Press the power-on button for one-two seconds. The display briefly shows the battery status and the moisture measurement mode, and then the upper line displays the name of the crop, or the label «SCALE UE», and the bottom line — the label «SELECT CROP».
Press the button shortly several times until the name of the grain to be controlled is displayed. In 3 seconds, the moisture meter automatically switches to the moisture measurement mode and then powers off in 5 seconds.
Use a scoop to pour the grain (cereal) without compacting it to the measuring chamber up to the edges to the full. Put the scoop aside.
Press the power-on button for one-two seconds and log in to the «MOISTURE MEASUREMENT» mode. Release the button. The display will highlight the name of the selected grain (cereal), and the digital result for measuring moisture content. The moisture meter powers off in 5 seconds automatically.
Pour the grain out of the measurement chamber. To make results more precise, repeat the procedure three times and calculate the average value.
1. Do not set the name of the grain (cereal) during repeated measurements conducted for the same crop.
2. Every time the (power-on) button is pressed, the measurement lasts for 5 seconds, followed by automatic power-off of the moisture meter.
3. If the moisture contents of the controlled grain (cereal) is below or above measurement range limits, the display highlights zeros or tens, respectively.